How do you add a strip lot to Accu Chek Inform 2?
From the main menu, press the right arrow button. Press Strip Lots. Press Add. Place the code key reader on a level surface, and slide the code key into the code key reader.
At what temperature should the Accu Chek Inform II strips be stored?
Use the test strips at temperatures between 61–95 °F (16–35 °C). Use the test strips between 10–80 % relative humidity. Humidity is the amount of dampness in the air. Store the test strips at temperatures between 39–86 °F (4–30 °C).
How often does the Accu Chek Inform II need to be charged?
The icon indicates the battery is charging. REMEMBER: Two levels of control must be run at least once every 24 hours and when opening a new vial of strips. If 2 lots of strips are in use, QC should be performed on both lots every 24 hours.
What is the proper procedure for placing the test strips?
Insert test strip into the test strip slot and the meter will turn on automatically. When the blood drop symbol flashes next to the test strip symbol, you are ready to perform a test. 2. Hold your finger to the tip of the strip until the yellow window is completely filled with blood.
What is the normal range for Accu Chek?
What should my blood glucose range be?
|Blood Glucose Levels||Fasting||Two hours after meals|
|Normal person without diabetes||70—99 mg/dl||< 140 mg/dl|
|ADA recommendation for diabetics patients||80—130 mg/dl||< 180 mg/dl|
What does it mean if you get a glucose result of Lo or RR Lo?
“RR LO” is an abbreviation for “reportable range, low limit,” which means the blood sugar is below normal. “CR LO” is an abbreviation for “critical range, low limit,” which means the blood sugar is critically below normal.
What is the normal range for Accu-Chek?
Where do you store the Accu-Chek Inform II glucose meter?
REAGENTS / MATERIALS: Accu-Chek Inform II Test Strips, Roche Diagnostics 1. Store at room temperature between 39-86 °F (4-30 °C).
When to use Accu-Chek inform 2 test strips?
The ACCU‑CHEK Inform II test strips are for use with the ACCU‑CHEK Inform II meter to quantitatively measure glucose (sugar) in venous whole blood, arterial whole blood, neonatal heelstick, or fresh capillary whole blood samples drawn from the fingertips as an aid in monitoring the effectiveness of glucose control.
How long are Accu Chek inform 2 controls good for?
•Controls are good for 90 days(3 months) from open date(Please date upon opening) Accu-Chek Inform II test strips •Strip must be used within 3 minutes following removal from test strip vial
What is the Accu-Chek inform 2 blood glucose test?
The ACCU-CHEK Inform II system provides healthcare professionals with an accurate, reliable, and easy-to-use blood glucose system to aid in monitoring the effectiveness of glucose control. The ACCU-CHEK Inform II meter and test strips provide a reliable, 5-second test with underdose detection,
How do you enter patient ID on Accu-Chek inform?
Enter the Patient ID by pressing the scan button on the upper right hand of the screen and positioning the scanning beam, located on the top of the underside of the meter, about 4 to 8 inches above the patient ID barcode. A beep indicates a successful scan of the patient ID. 4. The patient’s FIN number and name will display on the meter.